E443xB ESG Signal Generator Option UN8 or UN8/UN9, Software Patch
The Baseband Generator board with part number E4400-60070, serial number starting with 300, and version number A0110 contains wrong header information. This condition generates I/Q offset problems. To check if your E443xB ESG-D has the defect, access the installed board information on the display of the ESG. Using the front panel press Utility > Instrument Information > Installed Board Information.
To remedy the problem, download and run the self extracting executable Software Patch "BBG_rev.exe" available below.
|릴리스 날짜||버젼||버전 설명|
|2004-02-20||1.0||Self-extracting executable Software Patch.|
|Self-extracting executable Software Patch.|
- Click the Download button
- Save the file to your PC
- Activate the executable file
- Follow the installation prompts
- Does Not Require a License
Installs on: Instrument
- E4430B, E4431B, E4432B, E4433B, E4434B, E4435B, E4436B, E4437B with Option UN8 or UN8 UN9.