Test Cell Phones and Chargers with the Same Equipment
Notice to EU customers: This product is no longer available due to lack of compliance with EU RoHS directive 2011/65/EU. Keysight will continue service and support to the end of worldwide support life.
The DC Source 663xx series offers several features ideal for testing wireless and battery powered devices. Excellent voltage transient response ensures maximum test system throughput by minimizing device shutdowns resulting from significant voltage drops in the test wiring. A built-in advanced measurement system accurately measures battery current drains when the device operates in different modes and a unique feature enables cellular telephone manufacturers to detect permanently and intermittently open-sense wire connections.
The Keysight 66319B dual output mobile communications DC source provides two outputs for testing the main battery power and the battery charger circuitry, allowing customers to test phones and chargers with the same equipment. This saves you rack space (2 outputs in a ½ rack box) and eliminates the need for two separate power supplies.
R&D engineers can characterize current out of the battery and make tradeoffs in design that impact the current drain and battery life (by providing data logging – up to 1000 hours, CCDF statistical analysis, and waveform capture) , when the 66319B is used with the Keysight 14565B Device Characterization Software with Test Automation.
- Voltage: 0 to 15 V
- Current: 0 to 3 A
- Peak Current for up to 7 ms: 5 A
Programming Accuracy at 25°C ±5°C
- Voltage: 10 mV
- + Current: 1.33 mA
DC Readback Accuracy
- Voltage: 0.03% + 5 mV
- Current (-3 A to + 5 A): 0.2% + 0.5 mA
- Current (-1 A to + 1 A): 0.1% + 0.2mA
- Current (-20 mA to + 20 mA): 0.1% + 2.5 µA
- 40 mV drop, <20 µs transient response time
Interested in a 66319B?
Extend the Capabilities
Configuration Guides 2014.07.30
66300 Series Mobile Communications DC Power Sources
Application Notes 2017.11.30
Evaluate Battery Run-Down Performance of Mobile Devices
Application Notes 2021.05.18
Current Drain Analysis Enhances WLAN Network Card Design and Test
Application Notes 2021.05.30
Battery Drain Analysis for Low Power IoT Devices
Application Notes 2020.02.01
Evaluating Battery Run-Down with the N6781A or N6785A and the BV9200B
Technical Overviews 2021.11.16
Power Products Solutions
User Manuals 2014.10.31
66319B/D, 66321B/D User's Guide
View All Resources
66319D Dual Mobile Comm DC Source Battery Emulation, DVM
66309D Dual Mobile Communications DC Source w/ DVM
66309B Dual Mobile Communications DC Source
66321D Mobile Comm DC Source with Battery Emulation, DVM
66321B Mobile Comm DC Source w/ Battery Emulation