Notice to EU customers: This product is no longer available due to lack of compliance with EU RoHS directive 2011/65/EU. Keysight will continue service and support to the end of worldwide support life.
- voltage: 0 to 15 V
- current: 0 to 3 A
- peak current for up to 7 ms: 5 A
Programming accuracy at 25°C ±5°C (% of setting plus fixed)
- voltage: 0.05% + 10 mV
- +current: 0.05% + 1.33 mA
DC readback accuracy
- voltage: 0.03% + 5 mV
- current (-3 A to + 5 A): 0.2% + 0.5 mA
- current (-1 A to + 1 A): 0.1% + 0.2 mA
- current (-20 mA to + 20 mA): 0.1% + 2.5 µA
- 40 mV drop, <20 µs transient response time
Key Specifications apply to output 1 only
The Keysight Mobile Communications DC Source 663xx series offers several features ideal for testing wireless and battery powered devices. Excellent voltage transient response ensures maximum test-system throughput by minimizing device shutdowns due to significant voltage drops in the test wiring. Built-in advanced measurement system accurately measures battery current drains when the device operates in different modes, and a Keysight-developed feature enables cellular telephone manufacturers to detect permanently and intermittently open-sense wire connections.
The Keysight 66319D dual output mobile communications dc source provides two outputs for testing both the main battery power and the battery charger circuitry, allowing customers to test phones and chargers with the same equipment. This feature helps save on rack space (2 outputs in a ½ rack box) and eliminates the need for two separate power supplies. The 66319D dc source provides all of the capabilities of the 66319B with the addition of a built-in digital voltmeter.
Research and development engineers can characterize current out of the battery and make tradeoffs in design that impact the current drain and battery life by providing data logging (up to 1000 hours), CCDF statistical analysis, and waveform capture when the 66319D is used with the Keysight 14565B software.
Interested in a 66319D?
Extend the capabilities for your Dual Mobile Comm DC Source Battery Emulation, DVM
Featured Resources for 66319D Dual Mobile Comm DC Source Battery Emulation, DVM
Configuration Guides 2014.07.30
66300 Series Mobile Communications DC Power Sources
Application Notes 2017.11.30
Evaluate Battery Run-Down Performance of Mobile Devices
Application Notes 2021.05.18
Current Drain Analysis Enhances WLAN Network Card Design and Test
Technical Overviews 2017.12.01
Mobile Communications DC Sources Product Overview
User Manuals 2014.10.31
66319B/D, 66321B/D User's Guide
14565B Software and 66319B/D and 66321B/D Mobile Communications DC Sources – Product Fact Sheet
Technical Overviews 2022.04.13
Choosing the Right DC System Power Supply
View All Resources
66319B Dual Mobile Comm DC Source w/ Battery Emulation
66309B Dual Mobile Communications DC Source
66309D Dual Mobile Communications DC Source w/ DVM
66311B Mobile Communications DC Source, 15V, 3A
66321B Mobile Comm DC Source w/ Battery Emulation