Frequency coverage greater than 110 GHz, up to the limit determined solely by the broadband or banded millimeter-wave frequency extenders used in the system.
Supported on all PNA /PNA-X test set based millimeter-wave solutions
Supports broadband N5290A/91A with either 2- or 4-port configurations
Supports banded waveguide frequencies to above 110 GHz
Utilize the Cal All feature on the PNA/PNA-X for calibrated spectrum analysis measurements
DANL for a 500 to 750 GHz WR1.5 is about -106 dBm/Hz (typical)
Multichannel spectrum analysis at THz frequencies
Either N5261A/62A or N5292A millimeter-wave controller is required
The S93094A has all of the capability of the S93090xA spectrum analyzer application, with an upper frequency limit determined solely by the broadband or banded millimeter-wave frequency extenders used in the system. With this configuration, the analyzer’s internal receiver attenuators are not in the receiver paths, so they cannot be used to reduce signal levels. External attenuators may be required to avoid compressing the receivers in the millimeter-wave frequency extenders. When using the S93094A for millimeter-wave spectrum analysis, either N5261A/62A or N5292A millimeter-wave controller is required. When S93094A is installed on a standalone network analyzer, it will provide spectrum analysis up to the frequency limit of the instrument.