
Products + Services

Oscilloscopes + Analyzers
 Oscilloscopes
 Spectrum Analyzers (Signal Analyzers)
 Network Analyzers
 Logic Analyzers
 Protocol Analyzers and Exercisers
 Bit Error Ratio Testers
 Noise Figure Analyzers and Noise Sources
 HighSpeed Digitizers and Multichannel Data Acquisition Solutions
 AC Power Analyzers
 DC Power Analyzers
 Materials Test Equipment
 Device Current Waveform Analyzers
 Parameter and Device Analyzers, Curve Tracers
 Meters
 Generators, Sources + Power
 Software
 Wireless
 Modular Instruments
 Network Test
 Network Security + Visibility
 Additional Products
 Services
 All Products, Software, Services

Oscilloscopes + Analyzers

Solutions
 Industries
 Insights
 Resources
 Buy
 Support
Noise Figure Measurement Accuracy: The YFactor Method
1 Introduction
Why is noise figure important?
Noise figure is a key performance parameter in many RF systems. A low noise figure provides an improved signal/noise ratio for analog receivers and reduces bit error rate in digital receivers. As a parameter in a communications link budget, a lower receiver noise figure allows smaller antennas or lower transmitter power for the same system performance.
In a development laboratory, noise figure measurements are essential to verify new designs and support existing equipment.
In a production environment, lownoise receivers can now be manufactured with minimal need for adjustment. Even so, it is still necessary to measure noise figures to demonstrate that the product meets specifications.
Why is accuracy important?
Accurate noise figure measurements have significant financial benefits. For many products, a guaranteed low noise figure commands a premium price. This income can only be realized, however, if every unit manufactured can be shown to meet its specification.
Every measurement has limits of accuracy. If a premium product has a maximum specified noise figure of 2.0 dB, and the measurement accuracy is ± 0.5 dB, then only units that measure 1.5 dB or lower are marketable. On the other hand, if the accuracy is improved to ± 0.2 dB, all products measuring up to 1.8 dB could be sold at the premium price.
Customers need accurate noise figure measurements to confirm they are getting the performance they have paid for. Using the same example, an accuracy of ± 0.5 dB for measuring a product that a manufacturer has specified as ‘2.0 dB maximum’ would require the acceptance of units measuring as high as 2.5 dB. An improved accuracy of ± 0.2 dB sets the acceptance limit at 2.2 dB.
Speed of measurement is also an issue. Highvalue products favor accuracy; high[1]volume products favor speed. Due to the random nature of noise and the statistical aspects of measuring it, there is always a tradeoff between speed and accuracy.
To optimize the tradeoff, it is necessary to eliminate all avoidable errors and quantify the uncertainties that remain.
This application note demonstrates how to improve noise figure measurement accuracy by following a threestage process:
 Avoid mistakes when making measurements
 Minimize uncertainties wherever that is possible
 Quantify the uncertainties that remain
This application note covers the following topics:
 Fundamentals of noise figure measurement using the Yfactor method. (Chapter 2)
 Noise figure mistakes to avoid (Chapter 3)
 Measurement corrections to improve accuracy (Chapter 4)
 Calculation of the remaining uncertainties – including software tools (Chapter 5)
 Other techniques that can reduce uncertainties (Chapter 6)
 Checklist for improving accuracy (Chapter 7).
This application note is specific to instruments that use the Yfactor method for noise figure measurement. Various features of Keysight Technologies products are mentioned as illustrative examples of the newest generation of noise figure analyzers and noise sources. Other products, however, may be used with the techniques discussed in this document.
2 Noise Figure Measurement
This chapter outlines the fundamental features of the Yfactor measurement technique for noise figure. Many instruments use the Yfactor technique, including:
 Keysight XSeries NFA noise figure analyzers (Figure 11)
 Keysight XSeries signal analyzers with noise figure measurement application
 Keysight FieldFox handheld microwave analyzers with noise figure measurements (Figure 12)
 Other noise figure analyzers and spectrum analyzers with noise figure measurement personality
The equations developed in this chapter follow the internal calculation route of the Keysight Technologies, Inc. products. The calculation routes of other noise figure instruments that use the Yfactor method are inevitably similar.
This chapter departs from previous explanations of noise figure calculations by making extensive use of the noise temperature concept. Although noise temperature may be less familiar, it gives a truer picture of how the instruments actually work—and most importantly, how they apply corrections to improve accuracy.
2.1 Fundamentals
2.1.1 What is noise figure?
As explained in Keysight’s Fundamentals of RF and Microwave Noise Figure Measurements, the fundamental definition of noise figure F is the ratio of:
(signal/noise power ratio at the input of the device under test)
(signal/noise power ratio at the output of the device under test)
Or alternatively:
Noise figure represents the degradation in signal/noise ratio as the signal passes through a device. Since all devices add a finite amount of noise to the signal, F is always greater than 1. Although the quantity F in equation 21 has historically been called ‘noise figure’, that name is now more commonly reserved for the quantity NF, expressed in dB:
Keysight Technologies literature follows the contemporary convention that refers to the ratio F as ‘noise factor’, and uses ‘noise figure’ to refer only to the decibel quantity NF.