Confirm Your Country or Area
- Canada (English)
- Canada (Français)
- United Kingdom
- United States
- Hong Kong, China
- Switzerland (French)
- Switzerland (German)
Confirm your country or area to access relevant pricing, special offers, events, and contact information.
Products + Services
Oscilloscopes + Analyzers
- Spectrum Analyzers
- Network Analyzers
- Vector Signal Analyzers
- Logic Analyzers
- Protocol Analyzers + Exercisers
- Bit Error Ratio Testers
- Noise Figure Analyzers + Noise Sources
- High-Speed Digitizers + Multichannel Data Acquisition Solutions
- AC Power Analyzers
- DC Power Analyzers
- Dynamic Signal Analyzers, Materials Measurement
- Device Current Waveform Analyzers
- Parameter + Device Analyzers, Curve Tracer
- Generators, Sources + Power
- Modular Instruments
- Network Test
- Network Security + Visibility
- KeysightCare Service and Support
- Calibration Services
- Repair Services
- Technology Refresh Services
- Test as a Service (TaaS)
- Test Asset Management and Optimization
- Consulting Services
- Financial Services
- Infoline Support Portal
- Ixia Professional Services
- Training Services
- Used Equipment from Keysight
- All Services
- All Products, Software, Services
- Oscilloscopes + Analyzers
- Success Stories
It may surprise you to learn that there isn't a formal standard for how general-purpose measuring instruments are specified, or how those specifications are determined. Furthermore, when “calibration to manufacturer's specifications” is requested, a common assumption is that all calibration laboratories use a common method to qualify their calibration capability. Except for ISO/IEC17025 accredited calibration, this is not the case.
“Measurement adequacy” has often been judged by a relatively simple TAR (test accuracy ratio) approach, considering only the principal accuracy specification of the test equipment rather than all of the contributors to the total uncertainty.
The “Guide”, or GUM as it has become known, is a 100+ page document which recommends a methodology for qualifying test accuracy (or more correctly, the potential test inaccuracy) and was published in 1993. Although comprehensive in its treatment of the subject, it doesn't enjoy universal acclaim. Statisticians say it is critically flawed. Practicing engineers view it as too complicated and lengthy. It is, nevertheless, the first methodology to achieve formal, international consensus after decades of debate.
GUM Guide to the Expression of Uncertainty in Measurement – ISO/IEC Guide 98-3:2008
Visit http://www.iso.org to purchase this publication.
Free Version of GUM Guide to the Expression of Uncertainty in Measurement – OIML G 1-100:2008
A free version of the GUM is available from the International Organization of Legal Metrology. Download it by visiting http://www.oiml.org and search explicitly for “Evaluation of measurement data ‑ Guide to the expression of uncertainty in measurement”.
Free Practical Guides On Measurement Uncertainty
Before and after release of the GUM several national guides have been produced which aim to provide practical advice for those equipment users who are more likely to be at technician level rather than scientists.
EA-4/02 Expression of the Uncertainty of Measurement in Calibration
Released in December 1999, this is a comprehensive (79 pages) and authoritative publication (from Europe's official accreditation organization). Other than price (FREE), what makes this especially valuable is the number of examples that are used to illustrate the uncertainty budgeting process.
The first, widely acknowledged practical guide was undoubtedly the old British Calibration Service's document B3003 that addressed electrical calibrations and which subsequently became NAMAS NIS3003 and later renamed M3003. The latest 82-page version (edition 2, January 2007) is consistent with the GUM and includes examples from the non-electrical fields.