Refine the List
Tous les sujets
By Type of Content
- FAQ (2)
Par catégorie de produit
1-2 of 2
What is the possible cause of the difference when the permittivity measured with the 16453A is different from the value previously obtained?
Most likely cause of the measurement discrepancy or drift is insufficient electrode pressure due to improper pressure adjustment.
My 16099A test fixture adapter cannot be connected to the 16191A/16192A/16193A.
An old version of the 16099A cannot be connected to the 16191A/16192A/16193A. A modification is required.