N4916B De-emphasis Signal Converter with Optional Clock Multiplier
Refine the List
By Type of Content
1-10 of 10
|N4916B De-emphasis Signal Converter - Data Sheet
The N4916B de-emphasis signal converter enables R&D and test engineers to accurately characterize gigabit serial ports and channels. The clock doubler option is needed to analyze half-rate clock devices.
Data Sheet 2015-03-05
PDF 2.96 MB
|N4903B J-BERT - Getting Started Guide
Revision 11 (Released with N4903B Software 7.50).
Quick Start Guide 2013-11-01
|J-BERT N4903B High-Performance Serial BERT - Programming Guide
Released with N4903B Software 7.50. This document provides the information you need for programming the Serial BERT using the Keysight IO Libraries Suite. Familiarity with the Keysight IO Libraries Suite is instrumental in understanding remote programming of the Serial BERT. See the user documentation delivered with the Keysight IO Libraries Suite for information on how to use them.
Programming and Syntax Guide 2013-11-01
|J-BERT N4903B High-Performance Serial BERT - User's Guide
The Serial BERT can help you test the performance of components and systems for high-speed digital transmission equipment. An important parameter of a digital system is the rate at which errors occur in the system. The analyzer will compare the bits sent to your device with the bits received from your device. It will record the number of bit errors and display results such as the bit error ratio (BER). In addition, accumulated results can be analyzed to help you determine what caused the bit errors. (Version 7.50)
User Manual 2013-11-01
|Press Releases for N4916B
Press Releases for N4916B
Press Materials 2013-05-06
|N4916B De-Emphasis Signal Converter Getting Started Guide
Revision 7 (Released with N4916B Software 126.96.36.199). You only need a few minutes to get started with the N4916B. This Getting Started Brochure helps you to quickly install and check the instrument.
Quick Start Guide 2012-10-01
|N4916B De-Emphasis Signal Converter User's Guide
Revision 3 (Released with N4916B Software 188.8.131.52).
User Manual 2012-02-01
PDF 921 KB
|How to Pass Receiver Test According to PCI Express® 3.0 CEM Specification - Application Brief
This paper provides insight into the calibration method and tests, as well as the tools available. The biggest change between PCIe 2.x and rev. 3.0 is that RX test on cards will now be normative.
Application Note 2011-11-30
|N4915A-014 PCI Express 3.0® Calibration Channels - Data Sheet
The N4815A-014 PCI Express 3.0 calibration channels allow to generate receiver stress conditions according to PCIe 3.0 base specification rev 1.0 when used together with J-BERT N4903B and N4916B.
Data Sheet 2011-07-05
|Accurate Calibration of Receiver Stress Test Signals for PCI Express® Rev. 3.0 - Application Note
This paper describes the calibration of the receiver-stress signal according to the base specification of PCIe3. The calibration of the RX test signal is different from PCIe 2.0.
Application Note 2011-06-22